How are inconsistent values handled when using the Penguin RFA?
Helpful hints for handling inconsistent values.
Here are a few helpful hints for handling inconsistent values when using the Penguin RFA:
- Check to see if you are using the correct MultiPeg™ type for the implant system and implant platform size.
- Back out the MultiPeg™ and check for any seating interference from bone spicules or soft tissue tags. Reinsert the MultiPeg™ turning it in finger-tight and retake the ISQ value.
- If the same MultiPeg™ has been used more than 20 times, the threads may be worn and the MultiPeg™ should be replaced
- Ensure that the MultiPeg™ is fully seated, back it out and try to reinsert it until you can no longer rotate the test peg